Dome Shutter Control Issues
Wednesday, February 26, 2020 7:33 AM
The main shutter and dropout cannot be closed in the morning, after the observation.
The contact plate box, providing power to the dome dropleaf, shutter and fans had rotated, causing contact with the dome plate to be lost, even when the dome was in the home position. Re-aligned box and tightened screws. Problem solved.